NEW PRODUCT – Thermocouple Type-K Glass Braid Insulated – K
NEW PRODUCT – Thermocouple Type-K Glass Braid Insulated – K. Thermocouples are best used for measuring temperatures that can go above 100 degC. This is a bare wires bead-probe which can measure air or surface temperatures. Most inexpensive thermocouples have a vinyl covering which can melt at around 200 degC, this one uses a fiberglass braid so it can be used in high temperature measurements such as heaters and ovens.
Adafruit has had paid day off for voting for our team for years, if you need help getting that going for your organization, let us know – we can share how and why we did this as well as the good results. Here are some resources for voting by mail, voting in person, and some NY resources for our NY based teams as well. If there are additional resources to add, please let us know – adafruit.com/vote
Stop breadboarding and soldering – start making immediately! Adafruit’s Circuit Playground is jam-packed with LEDs, sensors, buttons, alligator clip pads and more. Build projects with Circuit Playground in a few minutes with the drag-and-drop MakeCode programming site, learn computer science using the CS Discoveries class on code.org, jump into CircuitPython to learn Python and hardware together, TinyGO, or even use the Arduino IDE. Circuit Playground Express is the newest and best Circuit Playground board, with support for CircuitPython, MakeCode, and Arduino. It has a powerful processor, 10 NeoPixels, mini speaker, InfraRed receive and transmit, two buttons, a switch, 14 alligator clip pads, and lots of sensors: capacitive touch, IR proximity, temperature, light, motion and sound. A whole wide world of electronics and coding is waiting for you, and it fits in the palm of your hand.
Get the only spam-free daily newsletter about wearables, running a "maker business", electronic tips and more! Subscribe at AdafruitDaily.com !
I wonder if this will work for the eternal question, “How hot is my soldering iron?”
will definitely work for that 🙂
Tip: sell it with a standard thermocouple connector.
I have a ‘k’ thermocouple thermometer, but the one problem it has is the wire is very brittle and can’t take much flexing. Replacement probes aren’t something you’ll find in your local walmart, so adding the standard connector should broaden the appeal of the product, imho.
s7, it would cost 2x as much – if you want a thermocouple with standard connectors, check omega.com
Hey ladyada, thanks for the plug in the tutorial! Nice looking breakouts.
signal7, like ladyada said the standard connectors do drive up the cost. I have some breakouts using “standard” connectors (ryanjmclaughlin.com/shop) and thermocouples, but I commonly get people asking why they are so expensive. Its just due to the thermocouple hardware. Sometimes it is just a bit overkill for a prototype, but these boards fill the gap.
I’ve actually used headless nails (small brads) to make ad-hoc connectors, with a rubber pencil eraser as a spacer in between and heat-shrink around the outside. Don’t use galvanized nails, btw. Of course, if you can afford a true Omega unit, you can probably afford the connectors too. Just sayin’