NEW PRODUCT – Analog Output K-Type Thermocouple Amplifier – AD8495 Breakout: Thermocouples are very sensitive, requiring a good amplifier with a cold-compensation reference. We have a couple digital thermocouple amplifiers in the shop already from Maxim. Now we’re happy to introduce an excellent analog-output amplifier. This is a very simple sensor to use, and if your microcontroller has analog input capability, you’ll be ready to go really fast!
The AD8495 K-type thermocouple amplifier from Analog Devices is so easy to use, we documented the whole thing on the back of the tiny PCB. Power the board with 3-18VDC and measure the output voltage on the OUT pin. You can easily convert the voltage to temperature with the following equation: Temperature = (Vout – 1.25) / 0.005 V. So for example, if the voltage is 1.5VDC, the temperature is (1.5 – 1.25) / 0.005 = 60°C
Each order comes with a 2 pin terminal block (for connecting to the thermocouple), a fully assembled PCB with the AD8495 + TLVH125 precision voltage reference, and pin header (to plug into any breadboard or perfboard). Goes great with our 1m K-type thermocouple (not included). Not for use with any other kind of thermocouple, K type only!
- Works with any K type thermocouple
- Will not work with any other kind of thermocouple other than K type
- Easy to use analog output
- Temp range with 5V power: -250°C to +750°C output (0 to 5VDC)
- Temp range with 3.3V power: -250°C to +410°C output (0 to 3.3VDC)
Have an amazing project to share? Join the SHOW-AND-TELL every Wednesday night at 7:30pm ET on Google+ Hangouts.
Join us every Wednesday night at 8pm ET for Ask an Engineer!
Learn resistor values with Mho’s Resistance or get the best electronics calculator for engineers “Circuit Playground” – Adafruit’s Apps!
Maker Business — Transforming Today’s Bad Jobs into Tomorrow’s Good Jobs
Wearables — Snap a picture
Electronics — To Y5V or not to Y5V?
Biohacking — Ticks are Spreading an Allergy to Meat
No comments yet.
Sorry, the comment form is closed at this time.