Thanks to Christer Weinigel for sharing this on his blog – be sure to read parts 2 and 3 for some further results.
What I really wanted was an active differential probe which should be able to look at a 500MHz signal at a couple of volts or so with decent fidelity. To do that I wanted a probe with about one GHz of usable bandwidth. I have an old Tektronix 11801B scope with two 20 GHz sampling heads with 50Ω single ended inputs that can handle signals which are 1.0V peak-to-peak so that’s what I wanted the probe output to be compatible with. The probe must attenuate the signal so to not overdrive the inputs and to make things easy to calculate I decided on letting the probe have a 1:10 attenuation.
Read part 1 here.
Join us every Wednesday night at 8pm ET for Ask an Engineer!
Maker Business — Transforming Today’s Bad Jobs into Tomorrow’s Good Jobs
Wearables — Snap a picture
Electronics — To Y5V or not to Y5V?
Biohacking — Ticks are Spreading an Allergy to Meat
No comments yet.
Sorry, the comment form is closed at this time.