Thanks to Christer Weinigel for sharing this on his blog – be sure to read parts 2 and 3 for some further results.
What I really wanted was an active differential probe which should be able to look at a 500MHz signal at a couple of volts or so with decent fidelity. To do that I wanted a probe with about one GHz of usable bandwidth. I have an old Tektronix 11801B scope with two 20 GHz sampling heads with 50Ω single ended inputs that can handle signals which are 1.0V peak-to-peak so that’s what I wanted the probe output to be compatible with. The probe must attenuate the signal so to not overdrive the inputs and to make things easy to calculate I decided on letting the probe have a 1:10 attenuation.
Read part 1 here.
Have an amazing project to share? Join the SHOW-AND-TELL every Wednesday night at 7:30pm ET on Google+ Hangouts.
Join us every Wednesday night at 8pm ET for Ask an Engineer!
Learn resistor values with Mho’s Resistance or get the best electronics calculator for engineers “Circuit Playground” – Adafruit’s Apps!
Maker Business — “ORANGE PI: MEETING WITH STEVEN ZHAO IN SHENZHEN”
Wearables — Putty in your hands
Electronics — Multimeter Bandwidth – AC Signal
Biohacking — Google Searching for “My Eyes Hurt” Peaked After Yesterday’s Eclipse
No comments yet.
Sorry, the comment form is closed at this time.